Subscribe to RSS
Automatic, log file-based process analysis of a clinical 1.5T MR scanner: a proof-of-concept studyArticle in several languages: English | deutsch
Purpose In light of the steadily increasing need for economical efficacy and capacity utilization it was the aim of this proof-of-concept work to implement an automated logfile-based analysis tool for MRI scanner utilization and to establish a process analysis. As a primary step, analyses of scanner and protocol utilization, parametrization of protocol processes, their durations, age dependency, and scan efficacy were to be tested.
Materials and Methods Logfiles were continuously extracted from a 1.5 T MR scanner (Philips Achieva) and automatically explored for relevant scan parameters. Parameters were extracted into a database and logically combined to protocol parameters. Visualization was achieved using PowerBI (Microsoft, USA). Data aggregation comprised a day-based and protocol-based strategy. In addition, age- and regional-based testing was performed. The frequency of protocol usage was evaluated and those protocols with frequent usage compared regarding efficacy to those rarely used.
Results After successful technical implementation, 3659 MR exams were available for further analysis. Out of a plethora of parameters, those relevant to the understanding of the scan process were identified. The initial results mirror the daily scanner usage and allow identifying, e. g., shortened scanner usage on Fridays or longer examination times in children. A scan efficacy of 69.6 ± 17.6 % excluding preparation process was identified as a parameter with high potential to be optimized in daily routine.
Conclusion The logfile-based analysis of MR scanner processes was successfully introduced and holds the promise to be extended into a comprehensive analytic tool for the analysis and optimization of scanner processes. In combination with other variables from the departmental or institutional infrastructure or patient-specific information such tool may be developed into a intelligent steering tool.
The automated log file analysis of MR-scanner processes was successfully introduced
The log file-analysis allows for a detailed analysis of scanner processes
From a log file-analysis, there is potential benefit to users, applications specialists and developers
Frydrychowicz A, Boppel T, Sieber V et al. Automatic, log file-based process analysis of a clinical 1.5T MR scanner: a proof-of-concept study. Fortschr Röntgenstr 2021; DOI: 10.1055/a-1346-0028
Key wordstechnical aspects - imaging sequences - health policy and practice - cost-effectiveness - MR-imaging
Received: 31 August 2020
Accepted: 22 December 2020
03 February 2021 (online)
© 2021. Thieme. All rights reserved.
Georg Thieme Verlag KG
Rüdigerstraße 14, 70469 Stuttgart, Germany
- 1 Deshmane A, Gulani V, Griswold MA. et al Parallel MR imaging. J Magn Reson Imaging 2012; 36: 55-72 . doi:10.1002/jmri.23639
- 2 Teichgraber UK, Gillessen C, Neumann F. [Methods of process management in radiology]. Rofo 2003; 175: 1627-1633 . doi:10.1055/s-2003-45331
- 3 O’Brien JJ, Stormann J, Roche K. et al Optimizing MRI Logistics: Focused Process Improvements Can Increase Throughput in an Academic Radiology Department. Am J Roentgenol 2017; 208: W38-W44 . doi:10.2214/Am J Roentgenol.16.16680
- 4 Chwang WB, Iv M, Smith J. et al Reducing Functional MR Imaging Acquisition Times by Optimizing Workflow. Radiographics 2017; 37: 316-322 . doi:10.1148/rg.2017160035
- 5 Boppel T, Frydrychowicz A, Schramm P. et al Introducing a logfile-based MR workflow analysis and efficacy analysis of compressed SENSE acceleration. Las Vegas, USA: ASNR; 2020
- 6 Kuhnert N, Lindenmayr O, Maier A. Classification of Body Regions Based on MRI Log Files. In: R KMWMB ed, 10th International Conference on Computer Recognition Systems CORES 2017 Springer; 2017
- 7 Gunn ML, Maki JH, Hall C. et al Improving MRI Scanner Utilization Using Modality Log Files. J Am Coll Radiol 2017; 14: 783-786 . doi:10.1016/j.jacr.2016.11.021
- 8 Andre J, Amthor T, Hall C. et al Radiological Assessment of Patient Motion Correlated with the Incidence of Repeat Sequences Documented by Log Files. Washington, USA: 54th Annual Meeting ASNR; 2016
- 9 Denck J, Landschutz W, Nairz K. et al Automated Billing Code Retrieval from MRI Scanner Log Data. J Digit Imaging 2019; 32: 1103-1111 . doi:10.1007/s10278-019-00241-z
- 10 Turney J. Six Sigma and Lean Six Sigma. Radiol Technol 2007; 79: 191-192
- 11 Bahensky JA, Roe J, Bolton R. Lean sigma – will it work for healthcare?. J Healthc Inf Manag 2005; 19: 39-44
- 12 Roth CJ, Boll DT, Wall LK. et al Evaluation of MRI acquisition workflow with lean six sigma method: case study of liver and knee examinations. Am J Roentgenol 2010; 195: W150-W156 . doi:10.2214/AJR.09.3678
- 13 Kruskal JB, Reedy A, Pascal L. et al Quality initiatives: lean approach to improving performance and efficiency in a radiology department. Radiographics 2012; 32: 573-587 . doi:10.1148/rg.322115128
- 14 Beker K, Garces-Descovich A, Mangosing J. et al Optimizing MRI Logistics: Prospective Analysis of Performance, Efficiency, and Patient Throughput. Am J Roentgenol 2017; 209: 836-844 . doi:10.2214/AJR.16.17698