Subscribe to RSS
Endoprotector: Protective box for safe endoscopy use during COVID-19 outbreak
Background and study aim Outbreak of COVID-19 began in Wuhan in December 2019 and has rapidly spread from China to the rest of the world. Although it is primarily transmitted by contact and droplets, aerosol-generating procedures also seem to carry the possibility of airborne viral transmission. As such, upper gastrointestinal endoscopy can be considered a risky procedure, with several position statements recommending the adoption of safe practices in endoscopic procedures. This article aims to describe an additional tool to protect healthcare personnel during endoscopy.
Methods A covering box made of acrylic plastic, named Endoprotector, was designed to cover the patient’s head, while taking into consideration movements performed by patients and healthcare staff during an endoscopic procedure.
Results A cough simulation using fluorescent dye confirmed the potential benefit of the box during endoscopy, thus protecting healthcare staff from air droplets. The feasibility and practicality of the box was also tested with patients during endoscopy.
Conclusion The reusable Endoprotector is easily and affordably manufactured and may reduce air droplets and airborne transmission of SARS-CoV-2 and other microorganisms during endoscopy.
Received: 12 May 2020
Accepted: 18 May 2020
02 June 2020 (online)
© 2020. Owner and Copyright ©
© Georg Thieme Verlag KG
Stuttgart · New York
- 1 Zhu N, Zhang D, Wang W. et al. A novel coronavirus from patients with pneumonia in China, 2019. N Engl J Med 2020; 382: 727-733
- 2 WHO Director-General’s opening remarks at the media briefing on COVID-19. 2020 Available from: https://www.who.int/dg/speeches/detail/who-director-general-s-opening-remarks-at-the-media-briefing-on-covid-19---11-march-2020
- 3 Coronavirus disease (COVID-19) Pandemic. Available from: https://www.who.int/emergencies/diseases/novel-coronavirus-2019
- 4 Modes of transmission of virus causing COVID-19. 2020 Available from: https://www.who.int/news-room/commentaries/detail/modes-of-transmission-of-virus-causing-covid-19-implications-for-ipc-precaution-recommendations
- 5 Johnston ER, Habib-Bein N, Dueker JM. et al. Risk of bacterial exposure to the endoscopist’s face during endoscopy. Gastrointest Endosc 2019; 89: 818-824
- 6 Repici A, Maselli R, Colombo M. et al. Coronavirus (COVID-19) outbreak: what the department of endoscopy should know. Gastrointest Endosc 2020; DOI: 10.1016/j.gie.2020.03.019.
- 7 Chiu PWY, Ng SC, Inoue H. et al. Practice of endoscopy during COVID-19 pandemic: position statements of the Asian Pacific Society for Digestive Endoscopy (APSDE-COVID statements). Gut 2020; 69: 991-996
- 8 ESGE and ESGENA Position Statement on gastrointestinal endoscopy and the COVID-19 pandemic. 2020 Available from: https://www.esge.com/esge-and-esgena-position-statement-on-gastrointestinal-endoscopy-and-the-covid-19-pandemic/
- 9 Canelli R, Connor CW, Gonzalez M. et al. Barrier enclosure during endotracheal Intubation. N Engl J Med 2020; 381: 1975-1958
- 10 Li R, Pei S, Chen B. et al. Substantial undocumented infection facilitates the rapid dissemination of novel coronavirus (SARS-CoV2). Science 2020; 368: 489-493
- 11 Bai Y, Yao L, Wei T. et al. Presumed asymptomatic carrier transmission of COVID-19. JAMA 2020; DOI: 10.1001/jama.2020.2565.
- 12 Zou L, Ruan F, Huang M. et al. SARS-CoV-2 Viral load in upper respiratory specimens of infected patients. N Engl J Med 2020; 382: 1177-1179
- 13 Huang C, Wang Y, Li X. et al. Clinical features of patients infected with 2019 novel coronavirus in Wuhan, China. Lancet 2020; 395: 497-506
- 14 Gu J, Han B, Wang J. COVID-19: Gastrointestinal manifestations and potential fecal–oral transmission. Gastroenterology 2020; 158: 1518-1519
- 15 van Doremalen N, Bushmaker T, Morris DH. et al. Aerosol and surface stability of SARS-CoV-2 as compared with SARS-CoV-1. N Engl J Med 2020; 382: 1564-1567