Eight Years’ Experience of Gastrostomy Tube Management
Submitted: 23 August 2001
Accepted after Revision: 19 July 2002
02. Dezember 2002 (online)
Background and Study Aims: Increasingly, patients fed by gastrostomy tube are surviving the lifespan of the device. Data are scarce concerning the factors affecting the longevity and failure of gastrostomy tubes or the criteria for selection of replacement devices which leads to cost-effective patient management. The aims of the study were: to set criteria for selection of replacement gastrostomy tubes; to determine the causes of gastrostomy tube failure, and the factors affecting device longevity; and to examine the effect of initiating an educational programme for caregivers on resource utilization in long-term enteral nutrition patients.
Materials and Methods: We analyzed the clinical gastrostomy tube database compiled prospectively over 8 years by the nutrition team at Ninewells Hospital, Dundee.
Results: For 363 gastrostomy tubes inserted in 304 patients (160 women; median age 71), the median duration of gastrostomy tube use was 138 days. The total follow-up was 294 patient-years. Death occurred before the first gastrostomy tube replacement in 48 % of patients, but 20 % resumed oral nutrition. Tube failure mechanisms were: dislodgment, 28 %; perishing of tube material, 25 %; tube-related Candida albicans infection, 16 %; leakage, 7 %; and unspecified, 7 %. Of the balloon tubes and gastrostomy buttons, 8 % needed early replacement due to dislodgment and/or leakage. The cost per day for replacement percutaneous endoscopic gastrostomy (PEG) was € 2.12, for balloon tubes it was € 0.62, and for gastrostomy buttons € 1.80. Despite an increasing PEG insertion rate throughout the study period, yearly referrals for PEG-related problems dropped by 30 % between 1997 to 1999, coinciding with the initiation of an educational programme for caregivers.
Conclusion: Tube longevity is mainly limited by the patient’s diagnosis and prognosis. The choice of replacement device should be based on clinical factors. The use of more durable materials in the manufacture of gastrostomy tubes may prolong tube life and reduce cost. Education of patients and caregivers by a multidisciplinary nutrition support team promotes independence and limits demand on the service.
- 1 Safadi B Y, Marks J M, Ponsky J L. Percutaneous endoscopic gastrostomy: an update. Endoscopy. 1998; 30 781-788
- 2 American Society for Gastrointestinal Endoscopy. Role of PEG/PEJ in enteral feeding. Gastrointest Endosc. 1998; 48 699-701
- 3 Gaudeger M. Twenty years of percutaneous endoscopic gastrostomy: origin and evolution of a concept and its expanded applications. Gastrointest Endosc. 1999; 50 879-883
- 4 Chowdhury M A, Batey R. Complications and outcome of percutaneous endoscopic gastrostomy in different patient groups. J Gastroenterol Hepatol. 1996; 11 835-839
- 5 Mathus-Vliegen L MH, Koning H. Percutaneous endoscopic gastrostomy and gastrojejunostomy: a critical reappraisal of patient selection, tube function and the feasibility of nutritional support during extended follow-up. Gastrointest Endosc. 1999; 50 746-754
- 6 Russell T R, Brotman M, Norris F. Percutaneous gastrostomy: a new simplified and cost-effective technique. Am J Surg. 1984; 184 132-137
- 7 Gauderer M WL, Picha G J, Izant R J Jr. The gastrostomy “button”. A simple, skin level non-refluxing device for long-term enteral feedings. J Pediatr Surg. 1984; 19 803-805
- 8 Gossner L, Keymling J, Hahn E G, Ell C. Antibiotic prophylaxis in percutaneous endoscopic gastrostomy (PEG): a prospective randomized clinical trial. Endoscopy. 1999; 31 119-124
- 9 Information and Statistics Division. Scottish Health Service costs. Edinburgh; NHS in Scotland 1997
- 10 Rimon E. The safety and feasibility of percutaneous endoscopic gastrostomy by a single physician. Endoscopy. 2001; 33 241-244
M. E. Koulentaki, M.D.
Gastroenterology Department, University Hospital Heraklion
PO Box 1352 · Crete · Greece
Fax: + 30-810-542085
eMail: [email protected]